09. Juni 2017
Neue AppNote: Nachweis von 3-MCPD und Glycidol in Speiseöl
Die GC/MS-Bestimmung von 3-MCPD und Glycidol in Speiseöl lässt sich basierend auf gängigen Methoden dank der Einbindung eines Eindampfschrittes äußerst effizient automatisieren
3-MCPD and Glycidol and especially their fatty acid
esters are process contaminants that are formed, for
example, when edible oils and fats are refined. At
least some of the above-mentioned substances are
classified as potential human carcinogens, a fact which
has prompted the introduction of rules and regulations
that specify tolerable daily intake values and maximum
levels in edible oils. Different analytical methods are
available for the determination of these compounds.
These methods follow two different strategies:
Direct determination or, more commonly, indirect
determination of the contaminants.
This AppNote describes a solution for fully automated determination of 3-MCPD and Glycidol in edible oils based on the reliable indirect method DGF C-VI 18 (10), similar to the ISO 18363-1 and AOCS Cd 29c-13 methods that are essentially identical. The edible oil sample is divided into two parts (assays A and B). Both are saponified using a Sodiumhydroxymethanol solution, but different quenching methods are used. In assay A, free Glycidol is converted to 3-MCPD using acidic quenching conditions in the presence of chloride. In contrast, for assay B, the quenching reagent is an acidic chloride free salt solution, in which free Glycidol is not converted into 3-MCPD. Following derivatization, the 3-MCPD amounts in both samples are determined by GC/MS as Phenylboronic acid (PBA) esters. Assay B is used to determine the amount of 3-MCPD in the sample while assay A provides the combined amounts of 3-MCPD and Glycidol. The amount of Glycidol is determined as the difference between the assay A and assay B results.
The work presented here involves an automated evaporation step as prescribed in the abovementioned official methods. This ensures that for most matrices, the required limits of detection can be reached using a single quadropole mass spectrometer (MSD). A further important aspect of the evaporation step is that it removes excess derivatization reagent, which could otherwise build up in the GC/MS system and influence system stability.
It is demonstrated that method ISO 18363-1, which is equal to both method AOCS Cd 29c-13 and method DGF C-VI 18 (10), can be automated using the GERSTEL MPS. The obtained results show good correlation with reference data. The excellent standard deviations achieved for the complete sample preparation and analysis workflow speak in favor of automation.
Fully Automated Determination of 3-MCPD and Glycidol in Edible Oils by GC/MS Based on the Commonly Used Methods ISO 18363-1, AOCS Cd 29c-13, and DGF C-VI 18 (10)